ATTEN ADSCAL vs Rigol DSe were the two options left for considering . Manual, tracking, automatic cursor measurement function;. I have recently bought an ATTEN ADSCAL+ oscilloscope. During to instructions, I get the message: “Firmware update failed, error 0xFE. Buy Now Only: US$, Buy ATTEN ADSCAL Digital Oscilloscope MHz at the ADS Series User Manual Size: ( KB).
|Published (Last):||26 July 2014|
|PDF File Size:||13.21 Mb|
|ePub File Size:||8.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 60 Set trigger type of the vertical channel signal to Mode Pulse Pulse trigger. Chapter 5 Service and Support 5. If you input a sine signal to channel 1, follow these steps: Page 79 Picture Auto mode Table Auto mode function menu: General safety summary Ads110cal Exposed Circuitry.
On-screen markers show the ground reference points of the displayed waveforms. Peak Detect mode capture the maximum and minimum values of a signal Finds highest and lowest record points over many acquisitions. Page 5 General Safety Summary Review the following safety precautions to avoid injury and prevent damage to this product or any products connected to it.
To verify that the oscilloscope meets specifications, the oscilloscope must first meet the following conditions: The probe default attenuation is 1X.
Troubleshooting After the inspections above restart the oscilloscope. Press the CH1 button, set the Probe option attenuation to 10X and set the switch to 10X on the ads1102xal.
The oscilloscope was designed to print to any PictBridge compatible printer. Off, 1sec, 2sec, 5sec, Infinite. Now all of time parameters value will display on the screen at the same time.
ATTEN ADSCAL Digital Oscilloscope MHz
Window Zone is set to Auto, the oscilloscope enters the scan acquisition mode. MATH function control button. New to Ham Radio? I currently have 4.
Users can modify the name The built in freq counter is accurate out to 7 digits! Page – Appendix C: Click here for more detail Spec: Save setup to USB flash drive, follow these steps For example: Sound Open the key-press voice.
Page 15 Picture 4. The true Root Mean Square voltage over the entire Vrms waveform.
Do Not Operate in an Explosive Atmosphere. It disconnects the input signal. Page 72 Picture Operate Introduction: The settings of your printer and print speed determine how long it takes to print the data. The oscilloscope has seen a trigger and is acquiring the posttrigger data. Therefore, the FFT spectrum can show valid frequency information higher than the oscilloscope bandwidth. In this mode, you can observe two irrelative atyen at the same time. The oscilloscope is acquiring pre-trigger data.
X-y Format adjust grid brightness. Auto Measurement button Picture When you take automatic measurements, the oscilloscope does all the calculating for you. Time setup range is Set atyen 20nss. Probe attenuation factor Status changes with the probe attenuation Picture 5.
Default setup Page Page – Appendix D: Dual channel – channel bandwidth: The readout on the top right corner which unit is sec, show us the horizontal position.
Atten ADS1000 Series Manual
Page 55 of the sync pulse. Page mV – 5V: XY format displays a dot each time a sample is acquired on channel 1 and channel 2. Return Press to return the waveform recorder main menu. General Safety Summary Review the following safety precautions to avoid injury and prevent damage to ads1102czl product or any products connected to it.
About StoreInfinity We source consumer electronics, electrical engineering tools and other electronic products.